When pressure sensors come into contact with hydrogen, this often results in difficulties. Unfortunately, I frequently notice in my own job that our customers have no idea of this prior to use. For this reason they often usually do not even reveal that their medium contains hydrogen. pressure gauge find out when we receive a complaint. But why do we get a complaint? Why is especially hydrogen so critical to pressure sensors?
A simple rule is that a large number of metals become brittle when subjected to hydrogen. Hydrogen atoms that diffuse into the metal grid change the effectiveness of the material. In the worst case, this results in cracks in the material. Or to put it simply, the pressure sensor starts leaking or is completely destroyed. In the case of pressure sensors, specifically thin membranes necessary for measuring pressure play a decisive role. Atomic hydrogen can not only penetrate the metal, but additionally pass through it by diffusion.
digital pressure gauge has several effects: When piezoresistive measuring principles are employed, it becomes critical if hydrogen reacts with the inner transmission medium or accumulates there. This initially only falsifies the measurement results, but can down the road also lead to complete destruction of the sensor in individual cases. The addition of hydrogen also changes the instrumentation characteristics of the resistance structures of the measurement bridge of thin-film sensors. Despite being reversible, this process results in a detuning of the bridge at least temporarily, resulting in a falsification of the measurement signal.
However, the consequences described above occur in various materials differently. Which means that the consequences of hydrogen could be lowered substantially by selecting suitable materials.
More info on our pressure sensors and on the subject of hydrogen can be found on the WIKA website. If you are using hydrogen because the medium, your contact will gladly recommend an optimum solution for your application.